Not without problems, but generally they work. You can also install this driver from another installation package. You can’t tell if you are “connected” to a device until you actual try to send and receive messages. You can download the latest version from: LabVIEW could read back some number of bytes but still throw a timeout error. During my research online, I have found out that prolific chips have kind of a bad reputation..
|Date Added:||11 April 2006|
|File Size:||32.92 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
So my first fault were the front panel settings, and my second one was a missing NULL modem.
GPIB interface is selected when the power supply is shipped from the factory. The first cable I tried did not work. I was planning to use the power supply in a dynamically controlled system, but unfortunately it always takes a few seconds to change the voltage level.
You can download the aiglent version from: If not found, the installer will abort, warn, or install the required component as appropriate.
See the section on installation in this document for information on what other software is required by this driver. LabVIEW could read back some number of bytes but still throw a timeout error.
Solved: Agilent EA – Page 2 – Discussion Forums – National Instruments
Thank you very much for the example for setting the default values myself. More Information For more information about this driver and other instrument drivers and software available from Agilent Technologies visit: However the close VI and the error query also seemed to have problems, which means that I would have to disable the error query in all of the other sub VIs of the driver Also connecting to a comport only means you are able to get as far as the com port itself.
So I began taking a closer look at the driver in Highlight Execution mode and started decoupling all the stuffs which made problems: Install The installation package for the driver is distributed as an MSI 2. You can also install this driver from another installation package. The GPIB interface is selected when the power supply is shipped from the factory. Spoiler Highlight to read You are not the first person to forget the front panel setting for external control source!
Agilent | Agilent E36xx IVI-COM Driver
The remote interface can be selected from the front-panel only. I guess the 8-ball is working fine today.
What kind of USB – serial device are you using? Everything worked the same.
Agilent Technologies / Keysight Technologies E3640A
If you plan to use the driver with. USB to RS converters are not all created equal. Are you saying by “connected” that you are able to send and receive commands when connected to a “built in” comm port of your PC using a program like Terra Term or Hyper Terminal, but not able to send and receive commands when using a 3rd party “RS to USB” converter? Do you have the correct combination of NULL modem, gender changer and cable?
Keysight Technologies / Agilent Technologies hpe364xa Power Supply
In my test at first I had problems with the cabling. Actually I have to admit that I am just using a gender changer.
As soon as the problem is solved I will post the solution everywhere I asked for help. This operation actually runs: Message 20 of NET framework late, you can use the file: Message 11 of Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. After that I had to remove the error queries from the Configure Output and the Configure Current Limit as well, which made these work too.
It is also available from: February 26, Version 1.